Industrial Oven,Drying Oven,Laboratory Oven,Hot air Oven
Industrial Oven,Drying Oven,it is a laboratory oven and hot air oven, can be used for powder coating,drying. The temperature up to 300℃ with good uniformity.
Internal capacity:100L to 1000L
Model no.: L-OV2 series
Temperature range:RT+10℃ ~200℃, RT+10℃ ~250℃,RT+10℃ ~300℃ for optainal
Controller:Single-point temperature controller, precise PID control
Unique structure and design, with good temperature uniformity
Specifications forIndustrial Oven,Drying Oven,Laboratory Oven,Hot air Oven
|Temp. range||Type 200:RT+10℃ ~200℃ Type 300:RT+10℃ ~300℃|
|Heating time||Ambient temperature~Highest temperature|
|RT+10℃ ~200℃≤40min,RT+10℃ ~200℃≤60min|
|Interior dimension (W*H*Dmm)||500*450*450||600*600*600||800*800*800||1000*1000*1000|
|Exterior dimension (W*H*Dmm)||1220*865*715||1320*1015*865||1520*1215*1065||1720*1415*1265|
|Temp control system||Balanced Temperature Control system (BTC system)|
|Enclosure||Colour coated sheet matel|
|Internal||Stainless steel SUS201|
|Insulation material||Glass wool|
|Heater||Nichrome Electronic Heater. Heater control mothed: PID + SSR contactless control|
|Controller||Single point digital controller|
|Running method||Fix operation,Single point control|
|Setting method||Button push setting|
|Display resoluation||Temperature:0.1℃ Time:0.1min|
|Control method||PID, Balanced Temperature Control|
|Power supply||AC220(10%)V,(500.5)Hz ,Single phase + E, 50Hz||AC380(10%)V,(500.5)Hz ,3 phase +N +E|
|Standard accessories||Cable hole(Φ50)1pc,Specimen holder 2 set,power cable 1pc|
|Safety devices||Leakage breaker,Interior over heat protector,Fan over heat protector,Phase error alarm|
|Remarks:1,Above data is measured under ambient temperature is 25℃, relative humidity is 85%RH, without specimen, and keep two hours constancy temperature.
2.Except the above specifications, all non-standard requirements can do according to customers special requirements.
3, Manufacturer has the right to change the above data without noticed.
External: Double galvanized metal steel sheet, with spray-paint electrostatic processing surface
- Internal: High quality stainless steel with mirror polishing
- Insulation: Rigid polyurethane foam and glass fiber
- Heater: Nickel chrome alloy
|Test area||Electrical control|
|-Adjustable build-in over temperature protection
-Additional over temperature protection
-over temperature limitation setup protection
|-Blower over heat protection
-Phase sequence, phase lose protection
-Earth leakage protection
-Load short circuit protection
2.1 Installation site conditions
Install this test chamber in a place which satisfied the following conditions
2.1.1 Ambient conditions
Ambient temperature rangeпјљ 5пЅћ35в„ѓпј›
Humidity range пјљВ В В В В В В В В В в‰¤85%RH
Atmospheric pressureпјљВ В В В 86пЅћ106KPaпј›
Air qualityпјљВ В В HavenвЂ™t high concentration dust and corrosive gas
Floor requirementsпјљВ В В Flat, level floor with good ventilation
For the purpose of future easily ventilation, operation and maintaining, please install the test chamber according to below space requirements:
It needs space more than 800mm from 4 sides of test chamber.
It must have sufficient access space to ensure the door can be opened. Please do not put any objects on the front side of door, as the hot steam may damage the objects.
2.2В Power supply
2.2.1В Power supply
Check main power supply for the equipment (the diameters of electrical wire, power switch etc.) comply with below requirements:
Frequency and fluctuation: 50В±0.5Hzпј›
2.2.2В Ground wire
Connect the test chamber to ground can avoid electric shock and reduce the noise caused by improper running.
2.2.3В Connect power supply cable
Connect the mains power supply cable of test chamber to the external electricity power supply (Check the phase consequence, current, and power consumpation clearly before connection).
3.1 Does your factory has the ability to design according to different specifications?
Yes, we can design the walk in chamber which exactly meet your test requirements, including temperature and humidity, weathering etc.
3.2 В In order to get the chamber which meet my requirements, which kind of requirements I should pay attention to?
In order to design theВ walk inВ chamber which exact meet your requirements, please let us those requirements at least:
Internal dimension, or specimen dimensions, temperature range, humidity range, ramp rate (if applicable).
3.3 What’s the lead time?
Normally, for a walk in chamber,the lead time is 2 month after deposit received.
3.4В What’s kind of payment method you accepted?
In our factory, the payment term is T/T 30% downpayment before production, and T/T 70% before shipment. If different payment method required, please contact with our sales.
3.5 What’s kind of price term I can work with you?
We accept EXW, FOB Guangdong, CIF destination port.