Temperature Humidity Vibration Test Chamber, it is an environmental chamber with temperature and humidity to work with a vibration table.
This chamber can come with different size of removable floors. Squre hole floor to work with the slip table, round hole floor to work with the head expander. The sizes can be done exactly same as per user’s requests. Except the removable floors with hole, a solid floor also will be supplied to enable the chamber work alone when no need to work with the vibration system.
Temperature Humidity Vibration Chamber, it is an environmental chamber with temperature and humidity to work with a shaker, including the slip table and head expander.
1,Independent control system: Realise the high temperature, low temperature, constant temperature and humidity, vibration test or a comprehensive test.
2,Adopt moving orbit design, easily realize the comprehensive test or a single test requirements.
3.Our vibration temperature humidity combined environmental chamber test systems can match with different brands of vibration table from the domestic and overseas. Labtek can provide customers with a variety of specifications and performance to integrate with their existing vibration table.
Temperature Humidity Vibration Test Chamber
Internal capacity: 1700L
Model no.: L-TS series
Temperature range:-70℃ ~180℃
Humidity range：20% – 98%RH
Vibration table (not included): Can work with a vertical & horizontal vibration table
Lift system is included
Vibration Temperature Humidity Combined Environmental Chamber
|Temperature range||Type 20: -20℃～+150℃, Type 40: -40℃～+150℃,Type 70: -70℃～+150℃|
|Temp fluctuation /deviation||±0.5℃ / ±2.0℃|
|Humidity range /deviation (For THS model only)||20%～98%RH / ±3.0%RH(>75%RH)，±5.0%RH(≤75%RH)|
|Heating time||‘-20℃→+150℃， -40℃→+150℃ ，-70℃→+150℃|
|Cooling time||+20℃→-20℃≤40min (No load, average cooling rate: 1℃/min) Ramp rate: 3℃/min, 5℃/min etc，can be done according to the requests，|
|+20℃→-40℃≤60min (No load, average cooling rate is: 1℃/min) Ramp rate: 3℃/min, 5℃/min etc ， can be done according to the requests.|
|+20℃→-70℃≤90min (No load, average cooling rate is: 1℃/min) Ramp rate: 3℃/min, 5℃/min etc, can be done according to the requests.|
|Displayer||5.7″ or 7″ touch screen programmable controller|
|Running method||Program running, fix running|
|Program capacity||120 patterns. Max. 999 segment max. per patterns. 99 times repeats|
|Display resoluation||Temp：0.1℃；Time：0.1min, Humidity: 0.1%RH|
|Control method||PID, Balanced Temperature Humidity Control|
|Water supply method||Water pump|
|Water quality||Deionized water or distilled water, resistance>500Ω·m|
|Auto-feeding||Auto-feeding available, feeding water pressure: 0.2~ 0.4MPa|
|Work with vertical shaker||Solid floor plug
One round hole to work with expender
With rail to move in and out the chamber when shaker no need.
|Work with vertical / horizontal shaker||Removable floors. With rails. With lift.|
Refrigeration system Refrigeration system front view
Front door open -70℃ testing
Left side view of the chamber Rear view of the chamber
2.1 Installation site conditions
Install this test chamber in a place which satisfied the following conditions
2.1.1 Ambient conditions
Ambient temperature range： 5～35℃；
Humidity range ： ≤85%RH
Atmospheric pressure： 86～106KPa；
Air quality： Haven’t high concentration dust and corrosive gas
Floor requirements： Flat, level floor with good ventilation
For the purpose of future easily ventilation, operation and maintaining, please install the test chamber according to below space requirements:
It needs space more than 800mm from 4 sides of test chamber.
It must have sufficient access space to ensure the door can be opened. Please do not put any objects on the front side of door, as the hot steam may damage the objects.
2.2 Power supply
2.2.1 Power supply
Check main power supply for the equipment (the diameters of electrical wire, power switch etc.) comply with below requirements:
Frequency and fluctuation: 50±0.5Hz；
2.2.2 Ground wire
Connect the test chamber to ground can avoid electric shock and reduce the noise caused by improper running.
2.2.3 Connect power supply cable
Connect the mains power supply cable of test chamber to the external electricity power supply (Check the phase consequence, current, and power consumpation clearly before connection).
3.1 Does your factory has the ability to design according to different specifications?
Yes, we can design the walk in chamber which exactly meet your test requirements, including temperature and humidity, weathering etc.
3.2 In order to get the chamber which meet my requirements, which kind of requirements I should pay attention to?
In order to design the walk in chamber which exact meet your requirements, please let us those requirements at least:
Internal dimension, or specimen dimensions, temperature range, humidity range, ramp rate (if applicable).
3.3 What’s the lead time?
Normally, for a walk in chamber,the lead time is 2 month after deposit received.
3.4 What’s kind of payment method you accepted?
In our factory, the payment term is T/T 30% downpayment before production, and T/T 70% before shipment. If different payment method required, please contact with our sales.
3.5 What’s kind of price term I can work with you?
We accept EXW, FOB Guangdong, CIF destination port.