Detailed installation guide for labtek salt spray test chamber
Once you received the salt spray test chamber, unwrap the chamber with care. In order to make the shipment easier, lots of components dismantled before shipment, now the user has to assemble those parts at first.
Firstly, take out the cover of the chamber, and put the chamber cover in another direction.
Secondly, lock the slot well and screw it.
Thirdly, insert the screws for holding arm. Different model with different size holding arm.
Fourthly, connect the external power supply, connect to the air supply, setting the pressure. Pressure setting method: please pull out the pressure adjustment button at first, increase the pressure by twisting this button clockwise, decrease the pressure by twisting this button anti-clockwise. Adjust the pressure within range 0.2 -0.4Mpa. The default setting from the factory is 0.2 – 0.4 MPa already. Push the button down after the pressure set well. If you want to start the test, press “Power on” button on the control panel to start test.
Labtek can produce different size of salt spray test chamber, contact our specialists now to get the professional proposal as per your requests.